STIRRED ICE BATH MODEL 813
he most used temperature for calibration is 0°C.
The normal way of creating 0°C is via a mixture of ice and water in a Dewar Flask.
However, this can give errors of up to 4°C because water is densest at 4°C and so as the ice melts the temperatures at the bottom of the flask can rise to 4°C.
In the design of the ice flask offered by Isothermal Technology Ltd., these problems have been eliminated by stirring the water/ice mixture and segregating the ice from the water in the measuring zone.
This stirred ice/water bath is designed and built according to National Laboratory recommendations.
Using demineralised water, accuracies of ±0.005K are obtainable. Typically the bath will last for 4 hours before recharging with ice.
The ice is contained around and below the compartment where up to 4 probes can be placed for calibration or referencing purposes.
An option permits a water triple point cell to be maintained within the stirred ice bath.
|Accuracy using demineralised water||0°C at ±0.005K|
|Capacity||8 litres (approx.)|
|Depth of Immersion||350mm|
|Accuracy using comparison techniques||±0.001°C|
|Power||50 Watts, 108-130 or 208-240Vac 50/60Hz|
|Dimensions||Height 580mm, Width 420mm (including handle), Depth 250mm, Weight 15kgs|
|Options||814/01B Copper Equalising Block, 814/02 Mercury Thermometer Support Kit, 814-06-02 Small Water Triple Point Cell Kit, 814-06-04 Large Water Triple Point Cell Kit.|
|How to order||813 Stirred Ice Bath. Please specify voltage required.|